Luminex IS 2.1_2.2 to xPONENT 3.1 Rev 2 Upgrade Instructions User Manual
Luminex IS Versions 2.1 and 2.2 Upgrade Instructions to xPONENT 3.1 SP1 - For In Vitro Diagnostic Use
LxFirmware Update Tool Dialog Box
6. Select the SD Device option button (item 6, above). Wait until the
top line of the Luminex XYP/SD Instrument Firmware Ver-
sion text box (item 4, above) displays the current version of the
Luminex SD system firmware and the second line displays its sta-
tus. This may take up to ten minutes. The example above may not
match the dialog box on your system.
7. If the information in the Luminex XYP/SD Instrument Firm-
ware Version text box (item 4, above) indicates that the firmware
is up-to-date, then click Close. No further action is required; con-
tinue with step 12. If an update version is available, continue with
8. Click Upgrade adjacent to the Movement Device text box. The
Confirm LXSD Firmware Download dialog box opens. Click
Yes to continue with the upgrade.
9. The Download In Progress dialog box opens displaying the
progress. Read the information presented on the dialog box.
When the download is finished, the Status text box displays
Download Complete. Click OK.
10. Turn off the power to the SD for 5 seconds, then turn it back on.
LXR Runtime Version
Luminex Analyzer Firmware Version
Movement Device Textbox
Luminex XYP/SD Instrument Firmware Version
Status of Upgrade
Movement Device / SD Device Selection Button