Luminex IS Version 2.3 User Manual
Luminex IS Software Manual for Version 2.3
PN 89-00002-00-110 Rev. B
Green text represents a successful system calibration,
verification command, acquisition, or maintenance functions.
Red text represents failed commands, errors, or warnings.
Black text represents normal processes and actions.
Yellow shading indicates that a detailed description about the
processes or actions is available. This color may vary depending
on your tool tip color system settings.
To view details of a message, double-click the shaded row. A
dialog box opens providing details. To clear the Message log,
right-click in the Message Log area, and click Clear on the menu.
The Acquisition Detail tab offers advanced batch sample monitoring
and “on the fly” data acquisition without templates. The primary
function is real-time monitoring of batch sampling during acquisition
through a display of sample bead statistics, histogram, and dot plot
During batch acquisition, bead statistics can be useful if batch errors
occur. For example, if samples are constantly failing due to
insufficient bead count, you can monitor whether the failure is due to
low bead concentration or if other assay problems are present.
The Acquisition Detail tab provides access to the Developer
Workbench (DWB) software features. Details about these features
are provided in the Luminex Developer Workbench Guide Version
2.3 documentation. You must have the DWB software installed to
enable these features.
Acquisition Detail tab features:
Batch Name and Description
Batch Data Area
Acquisition Detail Toolbar
Figure 11 identifies the major features of the Acquisition Detail tab.
Caution: Do not alter kit manufacturer’s predefined templates or
create alternative templates for off the shelf kits unless instructed
by the kit manufacturer.