Fluke 87 V User Manual
Smooth Feature (Power Up Option Only)
Table 7. MIN MAX Functions
MIN MAX Function
Enter MIN MAX recording mode. The Meter is locked in the range displayed before you
entered MIN MAX mode. (Select the desired measurement function and range before
entering MIN MAX.) The Meter beeps each time a new minimum or maximum value is
(while in MIN MAX mode)
Step through maximum (MAX), minimum (MIN), average (AVG) and present values.
PEAK MIN MAX
Model 87 only: Select 100 ms or 250
s response time. (The 250 µs response time is
indicated by p on the display.) Stored values are erased. The present and AVG
(average) values are not available when 250
s is selected.
Stop recording without erasing stored values. Press again to resume recording.
(hold for 1 second)
Exit MIN MAX mode. Stored values are erased. The Meter stays in the selected range.