LG QHW 15 User Manual
- 3 -
SECTION 3 INSTALLATION
3.1 GENERAL INSTALLATION
The scales should be sited in a location that will not degrade the accuracy.
Avoid extremes of temperature. Do not place in direct sunlight or near air conditioning
Avoid unsuitable tables. The tables or floor must be rigid and not vibrate. Do not
place near vibrating machinery.
Avoid unstable power sources. Do not use near large users of electricity such as
welding equipment or large motors.
Avoid high humidity that might cause condensation. Avoid direct contact with water.
Do not spray or immerse the scales in water.
Avoid air movement such as from fans or opening doors. Do not place near open
Keep the scales clean.
Do not stack material on the scales when they are not in use.
3.2 INSTALLATION of QHW SERIES
The QHW Series comes with a stainless steel platform packed separately. Place the
platform in the locating holes on the top cover. Do not press with excessive force as
this could damage the load cell inside.
Level the scale by adjusting the four feet. The scale should be adjusted such that the
bubble in the spirit level is in the centre of the level and the scale is supported by all
four feet, If the scale rocks readjust the feet.
Attach the mains cable to the connector on the bottom of the scale. The power switch
is located on the base near the front of the scale.