5 module 5 - s, Yes ren-n, No stb-n – NOSHOK 1950 Series Compact Smart System 5-Digit Process Meter User Manual
Page 11: No chc-n
6.5 MODULE 5 - S
The Serial Setup Parameters are only active when the optional RS232 or RS485 serial communications module is installed in the meter.
button or user input manual reset, serial reset command or meter power
cycle. When the user input or
button is activated (momentary action),
the corresponding “on” output is reset immediately and remains off until the
trigger point is crossed again. (Previously latched alarms will be off if power
up Display Value is lower than setpoint value.)
= Latch with delay reset action; This action latches the output on at the
trigger point per the Setpoint Action shown in Setpoint Output Figures. Latch
means that the output can only be turned off by the front panel
or user input manual reset, serial reset command or meter power cycle. When
the user input or
button is activated (momentary action), the meter
delays the event until the corresponding “on” output crosses the trigger off
point. (Previously latched outputs are off if power up Display Value is lower
than setpoint value. During a power cycle, the meter erases a previous
reset if it is not activated at power up.)
Setpoint Output Reset Actions
This parameter enables the
button or user input to reset the output when
the display is reset.
Note: For this parameter to operate, the
button or User Input being used
must be set to
and the Input value must be displayed. If these conditions are
not met, the output will not reset.
OUTPUT RESET WITH DISPLAY RESET
, the output is disabled (after a power up) until the trigger point is
crossed. Once the output is on, the output operates normally per the Setpoint
Action and Output Reset action.
This parameter enables the backlight 1950 to switch the backlight color when
the output state changes. This parameter is only active for the backlight version.
CHANGE DISPLAY COLOR w/OUTPUT STATE