# Kds ezflow 2030, Operation procedures – KD Scientific EZFLOW 2030 User Manual

## Page 7

6

KDS EZFlow 2030

Operation Procedures:

(Continued)

SET INFUSION RATE – SIMPLIFIED METHOD (WITHOUT RULER)

To determine the infusion rate of the syringe without

1)

Load the empty syringe on the pump as instruction (refer to previous page

2)

Note the reading of the pump’s cursor location. (E.g. 6mm on the pump scale)

3)

Move the syringe by holding the syringe driver head by 1 ml according to the

reading on the syringe.

4)

Note the reading of the pump’s cursor location. (E.g. 10 mm on the pump scale).

5)

Calculate the difference on the pump’s scale reading.

(E.g. 10 mm - 6 mm = 4 mm). The difference is the conversion factor between mm

and ml of the current loaded syringe per ml. (E.g. 4 mm - 1 ml, the conversion

factor is 4 mm/ml)

6)

Set the desired infusion rate by

Formula : Conversion factor x desired ml/hr rate.

(E.g. If the desired driving rate is 10 ml/hr, then the equivalent rate on the pump

is 4 mm/ml x 10ml/hr - 40 mm/hr. Therefore, set the driving rate to be: 4 0 on

the pump)