Intek 100 User Manual
- 3 -
MODEL100 REV D.wpd
Check the transducer maximum temperature rating — do not operate a transducer at or subject
it to a temperature above its specified limit.
Keep moisture out of the electronic enclosure and sensor junction box. Once cable
connections are made in the junction box, make sure the lid is tightly closed. Seal conduit
lines if they can become wet inside.
Keep transducer wetted surfaces clean and free of permanent layer build-up.
Do not exceed pressure limits of the tube or fittings.
Maintain a thermally stable environment (short-term) for the transducer and adjacent line.
(See SECTION 2 — INSTALLATION.)
These instructions cover installation, calibration and maintenance of Rheotherm meters in standard
configurations. Any special information pertaining to your unit is covered under CUSTOM
INFORMATION (SECTION 6). Time should be taken to carefully read these instructions prior to
installation of the equipment. Should any questions arise or problems occur, call Intek for immediate