Unpacking useful facts commissioning – IKA KS 130 control User Manual
Please unpack the equipment carefully and check for any dama-
ges. It is important that any damages which may have arisen
during transport are ascertained when unpacking. If applicable a
fact report must be sent immediately (post, rail or forwarder).
The delivery scope covers:
A KS 130 basic or KS 130 control, for attachment screws, a
single-head spanner, a connection cable and operating instruc-
With the purchase of this device, you have acquired a high-qua-
lity product. The design of the unit and its special shape ensure
ease of handing and problem-free work. The materials used and
an exact designation of them will significantly facilitate and sim-
plify recycling and reuse of parts.
The speed-controlled external rotor asynchronous motor allows
for infinitie speed adjustment in the range from 0 to 800 rpm.
Electronic motor control holds the set speed constant even if the
weight of the material on the surface increases. The heat given
off by the motor can be used to heat up the holding surface for
the agitation containers.
The Control variant of the device is equipped with a locking devi-
ce. This makes it possible to fasten the agitation table in a defi-
ned position. By removing the rubber feet, the device can be
connected or screwed in place in a defined position or with a
system located next to it. Make certain that the maximum depth
of 5 mm for inserting the screws is not exceeded.
The Control variant of the device can be delivered with a reverse
operating mode (clockwise / counterclockwise motion) for the
vibration table on request. However, the reverse operation mode
can only be used by means of the serial interface in combination
with a PC (for example with labworldsoft). If the device is not
operated on a serial interface, it behaves like a Control device
without reverse operation mode.
It is easy to exchange the various attachments.
The Control device variant is equipped with a serial interface that
makes it possible to control the device through the PC (for exam-
ple with labworldsoft.
Check whether the voltage specified on the type plate matches
the mains voltage available. The power socket used must be ear-
thed (protective earth conductor contact). If these conditions are
met, the device is ready to operate after plugging in the mains
plug. If these procedures are not followed, safe operation cannot
be guaranteed and/or the equipment may be damaged.
Observe the ambient conditions (temperature, humidity, etc.)
listed under Technical Data.
When you turn on the basic device variant, the turn button for the
time setting must be in the middle setting (not activated).
If the mounting surface is not even, you can improve the safety
level of the device in reference to how it stands with the adju-
stable feet. To do this, turn the appropriate device foot down-
ward with the wrench (included with delivery) until the device is
standing securely on the surface.
To prevent the foot of
the device from coming
loose by itself, hold
down the foot with one
hand and tighten the
oppsite hexagonal nut
with the other hand
until it is finger tight.