Cleaning, Maintenance -- testing and inspection – ZOLL X Series Monitor Defibrillator Rev A User Manual
9650-000840-01 Rev. A
ll Battery Pack
Clean the SurePower II Battery Pack with a soft damp cloth and mild soap.
Do not immerse or set the SurePower II Battery Pack in fluids of any kind. Do not use alcohol
or ketones to clean the SurePower II Battery Pack.
Maintenance -- Testing and Inspection
Perform a battery check, which verifies that the battery is fully charged and fault free, as part of
any regularly scheduled unit test. Press the battery’s Display button to perform the battery
As part of your defibrillator/monitor maintenance program, verify that the battery’s LEDs work
(press the Display button to test LEDs) -- if the LEDs do not work, remove the battery from
service. Also check the SurePower II Battery Pack casing for cracks, damaged contacts and
sockets, or other damage -- if the battery pack is cracked or damaged, remove it from service.
SurePower II Battery Pack
This is the Recalibration icon.
This LED lights for approximately 10 seconds (after you
press and release the Display button) if the battery needs to
be calibrated. If the Recalibration LED lights, the runtime
indicator will not display run time for that battery. For best
performance of the battery, you should recalibrate the
battery as soon as possible.
To manually recalibrate the SurePower II Battery Pack, you
can insert the battery into the SurePower Charger Station
(using the SurePower
II Battery Adaptor
) and perform a
(for more information, see the ZOLL
SurePower Charger Station Operator’s Guide
). After you
recalibrate the battery, the Recalibration LED will only flash
momentarily when you press the Display button.
This is the Fault Icon.
If the SurePower II Battery Pack detects a battery fault, this
LED lights for approximately 10 seconds after you press the
If the Fault LED lights, the runtime indicator will not display
run time for that battery.
See the following section for more information on battery