6 crossarm attachment, 7 enclosure attachment, 1 enclosure mounting to tripod mast – Campbell Scientific CM110, CM115, CM120 Tripod Installation User Manual
Page 26: Crossarm attachment, Enclosure attachment, Enclosure mounting to tripod mast, 13. cm200-series crossarm
Tripod Installation Manual Models CM110, CM115, CM120
6.1.6 Crossarm Attachment
Attach the CM202 (2 ft, 0.6 m), CM204 (4 ft, 1.2 m), or CM206 (6 ft, 1.8 m)
crossarm to the tripod mast as shown in FIGURE 6-13. For wind sensors, the
crossarm should be approximately 103 inches above the ground for a 3 m
mounting height, or 64 inches for a 2 m mounting height. Typically the
crossarm is oriented east/west for wind sensors, north/south for pyranometers.
FIGURE 6-13. CM200-series crossarm
6.1.7 Enclosure Attachment
The ENC10/12, ENC12/14, ENC14/16, and ENC16/18 enclosures can be
ordered with mounting brackets for the CM100-series tripods. All enclosure
models can be mounted to the tripod mast (above the legs) with the –MM Mast
Mount bracket option. All enclosure models except the ENC16/18 can be
mounted to the tripod base and leg with the –LM Leg Mount bracket option.
Two enclosures with the –LM brackets can be mounted in a “back to back”
188.8.131.52 Enclosure Mounting to Tripod Mast
An enclosure ordered with the –MM bracket has a three-piece top and bottom
brackets with a U-bolt for each bracket.
Attach an enclosure with the –MM mounting bracket to the tripod mast as
Remove the U-bolts, washers, and nuts from the brackets.
Position the enclosure against the tripod’s mast (north side recommended).
Install the U-bolts, flat washers, lock washers, and nuts. Tighten the nuts
until the lock washers are compressed.