Intek RheoVac DR User Manual
- 19 -
HARDWARE AND SOFTWARE MODIFICATIONS AND UPGRADES
Note: The workgroup setting of the identification tab in Network Neighborhood Properties must be set
to WORKGROUP which is the instrument default setting. A Cat5 crossover network cable is required
for proper Ethernet network communication to the RheoVac DR instrument.
Normally, updating the main software residing on the RheoVac DR should be done at the factory.
However, software improvements or other factors may necessitate a user installation or upgrade of a new
main program. When this is required, instructions and software, as applicable, will be supplied.
If it is necessary to replace a probe, a new probe and diskette (with probe-specific calibration data), can
be ordered from Intek.