IKA C 200 User Manual
Switch the calorimeter on at ON/OFF switch
(back of appliance). The appliance is now in
Press ON (F1) to work with the appliance. The
start screen will appear.
The operating console features the following ele-
Display elements during operation:
Status line: shows the current status of the
Footer: shows what the function keys cur-
Exception: a progress bar is displayed during
automatic measuring runs.
Reading value: shows the current tempera-
ture increase in minute intervals during the
measurement or the measurement result after
Current temperature value: shows the cur-
rent temperature in second intervals at the
sensor in the inner vessel.
Function keys F1, F2, F3: what these keys
do depends on the operating state of the ap-
pliance. The footer of the display shows what
the function keys currently do.
Die Fußzeile des Displays zeigt die aktuelle Be-
legung der Funktionstasten an.
Numerical keypad: this is used to enter digits
and the decimal point into command lines.
Delete button: this key is used to delete the
last character entered.
Display elements in menu mode:
If you press the Menu (F3) key, a menu will ap-
pear in the display which allows you to enter set-
tings. There are six submenus:
UNIT OF MEASUREMENT
Pressing the DOWN (F2) key selects the menu
line below. Double clicking (F2) allows you to
switch between UP and DOWN. The current di-
rection of movement is displayed in the footer.
BACK (F3) allows you to exit a menu without
conﬁ rming new settings.
Selection menu (e.g. combustible crucible)
4.3 Switching on the system
4.4 Display and operating elements