3 interval, 4 amplitude, 5 start / stop – FRITSCH ANALYSETTE 18 User Manual
Page 27: 6 continue, 7 store, 8 load, Using the device
The interval time is set in seconds in the set point / actual time display.
Display 00 = continuous operation without interruption. Display 01 - 99 =
sieving time in seconds, interruption 1 second. After pressing "Start", the
display switches to the actual value display, which shows the remaining
The amplitude set point can be set in steps of 0.1 mm up to 2.0 mm.
After pressing "Start", the display switches to the actual value display,
which shows the measured amplitude value during sieving.
5.5.5 Start / Stop
This button starts the machine. The remaining sieving time is visible in
the set point / actual value display. During the sieving process the input
buttons for the set point are disabled. After pressing "Stop" the input
buttons for the set point are enabled.
This button can be used to continue a sieving process after "Stop" has
been pressed (this can only be used when the sieving time has been pre-
viously been set).
This button can be used to store 10 set points (time, interval, amplitude)
and will not be affected by power failures.
Enter the set point and press the "Store" button. Enter the memory loca-
tion (01-10) in the time display field (which flashes) and press "Store"
again to save the values.
This button can be used to recall the desired memory location (01-10).
Select the desired memory location with the time adjustment buttons.
The sieving process can be performed by pressing the "Load" button
Using the device
- 27 -