BNC SAM 935 Portable Gamma Spectroscopy System User Manual
SAM 935™ Instruction Manual
If you know what the value should be, press F3 to manually edit the value:
When finished, press E
to save the new value.
To perform the dose rate calibration, press F2. The SAM 935 will tell you that you need a Cs137
source of known activity and date to do this:
Enter the activity of the calibration standard at the
next screen and press E
Enter the standard’s calibration date at the next
screen and press E
You will be prompted to place the Cs137 standard 20
centimeters (7.87 in.) away from the SAM 935’s detector. Press OK (F2) when you have done so.
You will see a spectrum resolve on screen as the calibration is taken.
Contact your supplier if needed to get the value to
be entered manually here.